86-0311-68015506 ; 86-15032065295
Specific solutions to prosthesis leg for house failure
The specific solutions of prosthesis leg for house failure are as follows
1. It's more and more inappropriate to accept the prosthesis leg for house. Most of them are caused by atrophy of the stump or thinning of the lining material after a long time of use. Properly increasing the thickness of the stump socks can basically eliminate the fault, but the best solution is to replace the socket.
2. There is noise at the joint of pipe fittings. In case of noise, you can first check whether the fixed screw at the connection of the pipe fitting is loose. In many cases, because the contact surface of the fixing screw is too small and the strength is not enough, after a period of use, the contact is not in place. After fixing the fixed screw with a special screwdriver, it can be basically eliminated.
3. When walking, it takes a lot of effort or people rush forward. This condition is usually caused by changes in the heel height of the shoe. If the height is lower than the original heel, it will be hard to step. If the height is higher than the original, it will lead to the situation of walking forward. The temporary solution is to put a little something under the heel or forefoot of the foot. However, we should remind everyone that hard objects should not be padded to avoid damaging the foot board. The cushion is only temporary adjustment, and wearing suitable high-heeled shoes is the key.
The perfect prosthesis leg for house should be able to compensate the ankle joint and foot movement of normal people as far as possible. Because the foot of artificial limb needs to be able to bear the weight of durability, the price of components should be appropriate, the weight should be light, and the appearance should be good. Therefore, only the intelligent prosthesis on the market is closer to the ankle joint and foot movement of normal people. However, it is difficult to research and develop intelligent leg prosthesis technology, and the material is expensive, which leads to the high price of intelligent leg prosthesis. Therefore, the small editor today is mainly to explain to you about the energy storage foot with affordable price and more comprehensive functions.
Energy storage foot is a kind of high-performance artificial foot suitable for young and middle-aged people. It can not only meet the needs of daily walking and physical labor of amputees, but also meet the needs of patients with high activity intensity. The energy storage foot can save energy during the support period and release energy when the patient is exercising, so that the patient can reduce the physical consumption when exercising. The structural feature of this kind of energy storage foot is that the range of ankle dorsiflexion activity area is larger than that of common false feet, and the foot parts have energy storage effect and can also rotate inside and outside. Because of the need to consider the structural principle and weight factors, all the energy storage feet on the market are made of carbon fiber materials.